Do you need to detect ferrous wear particles in machinery? Are you working remotely, onboard a vessel, or onsite in a less than ideal environment? Ferrous wear meters allow users to measure ferrous wear debris in oil samples for condition monitoring in both marine engines and lubricated machinery.
The Parker Ferrous Wear Meter (FWM) is a simple, easy-to-use instrument that provides accurate, reliable detection of metal particles in oil samples taken from a variety of types of machinery, including used cylinder scrape down oil, gearboxes and bearings. Monitoring of ferrous wear levels with your oil samples on an ongoing basis allows any deterioration in machine condition to be quickly spotted and corrective action taken, saving you downtime and money, and preventing more serious damage occurring. This versatile unit is ideal for testing and analysing oil samples on-site, on-board or in remote locations where full laboratory analysis is not possible.
The FWM is constructed using a sophisticated magnetometer adapted for field applications. A 5-millilitre test tube, filled with the sample, is placed directly in the hole in the instrument and its metallic content, in PPM, is displayed on the screen in less than 2 seconds.
By taking ferrous wear measurements over time, any increase in wear levels can be monitored and appropriate actions taken to mitigate any damage. Machinery degradation can be observed as it happens and machines serviced as and when they need to be, rather than on a time or hours of operation basis, saving costs and manpower.
Each ferrous wear meter is supplied with three check standards to verify the product on a regular basis. In this example,
the 972 - 922 PPM ferrous content sample is being used.
As per the on-screen instructions, insert the sample into the FWM for approximately one second, then remove. The result will be displayed on the screen for comparison against the check standard.
Begin by agitating your oil sample for approximately 30 seconds. Fill a 5-millilitre test tube to within 5-millimetres of the top with your oil sample. For each test, use a new test tube.
The results will display shortly after removal.
Watch this step by step video for complete instructions:
Please visit our Hydraulic and Industrial Process Filtration Division EMEA for additional information.
This post was contributed by the Hydraulic Filtration Team.